IEC Standards for the Safety and. Efficacy of Medical Electrical Equipment . Introduction. It might, perhaps, be a little too self-evident to say that there is no. The new IEC standard for the in-service and post repair testing of electromedical devices introduces new requirements. These and their. STANDARD. IEC. CEI. NORME. INTERNATIONALE. First edition. Première édition. Medical electrical equipment –. Recurrent test and test after.
|Published (Last):||27 October 2005|
|PDF File Size:||4.79 Mb|
|ePub File Size:||6.60 Mb|
|Price:||Free* [*Free Regsitration Required]|
The test methods are:.
IEC | IEC Webstore
Published on September 7, Routine testing, also referred to as PPM, preventive product maintenance. Each part of the ice device market has different electrical safety testing requirements.
The requirement for safety testing medical electronic ME equipment is regarded as essential to ensure that apparatus does not pose any danger to users or patients.
Ultrasound Systems and Probes If the buzz at the recent RSNA annual meeting is any indication of market trends, the ultrasound sector is hotter…. The test methods are: Figures 1, 2, and 3 provide a schematic representation of the equipment leakage test on Class I grounded ME equipment. Once a medical device enters into service, a number of potential test scenarios arise, including:.
Although the onus will remain on the manufacturers of medical devices to advise on appropriate tests for their equipment, the new standard will clearly have a significant impact on medical service companies and clinical engineering, EBME, medical physics, and other technical departments. IEC Medical Electrical Equipment—recurrent test and test after repair of medical electrical equipment—defines the requirements of ensuring the in-service electrical safety of EM equipment and systems.
As a type-testing standard, the current IEC does not provide any guidance to standardizing test requirements once an item of ME equipment has passed the design phase.
If the buzz at the recent RSNA annual meeting is any indication of market trends, the ultrasound sector is hotter…. Stay aware of moving parts, open conductors, live components, heat, etc.
IEC Ground Bond Test The ground bond test proves the integrity of the 26353 connection between the ground conductor and any metal conductive parts, which may become live in fault situations with Class I medical devices.
An Introduction to IEC 62353
If money talks, then a lot can be said about the state of the healthcare technology management field. Ensure that the operator of safety test equipment is properly trained on both the safety analyzer and the device under test to prevent unneccessary danger during the safety test. The healthcare market is technology driven and with a global demand for increased patient safety, there has never been a more prudent time to demand more from your test equipment.
The assumption was made that higher currents could best detect any damaged conductors present. Another disadvantage is that secondary ground connections will produce a lower reading, thus potentially allowing faulty equipment to pass the test.
Please provide your name and email to continue. Acceptance testing is often not limited to electrical safety tests, with some basic iecc tests being applied to verify correct performance.
This standard is not applicable to the assembly of ME systems. Contact resistance can influence 62533 readings and cause unnecessary failures of the device under test. For the purpose of this standard, the application of measuring methods is independent of the edition according to which the ME equipment or ME system is designed. The disadvantages of using the alternative method are that measurements cannot be compared with previous IEC tests, and those active parts of the circuitry that require mains potential between live and neutral cannot be tested for 63253 leakage.
Research has shown that it is current rather than voltage that is the source of electricity-related injuries and deaths. In addition, this measurement is only taken in a single polarity and is similar to a dielectric test at mains potential using a current limited mains frequency supply.
But even when the need for safety testing is recognized, other factors come into play. The differential leakage measurement method is also only able to measure the AC component. The following test report forms are related: As such, consideration should be given to the test current.
Learning from close calls and hazardous conditions explores guidance for healthcare organizations and…. Otherwise conformity to applicable requirements should be assessed and verified, before the tests of this standard are performed. This second edition cancels and replaces the first edition of IEC published in All maintenance, inspection, servicing, and repair done in accordance with the manufacturer’s instructions maintain the conformity to the standard used for the design of the equipment.
Do you need a multi-user copy? The alternative method is similar to a dielectric strength test at mains potential, using a current limited voltage source at mains frequency. You can find in-depth information on IECincluding in-service test requirements in this part of 26353 free guide. Measurements can also be influenced by the presence of magnetic fields—the principle of measuring differential current—and measurements must be done in both directions to identify the worst-case scenario.
6235 are called leakage currents.
Safety testing of medical devices: IEC 62353 explained.
For example, the test limit for a Class I device including a power cable not exceeding 3 meters ice mV. This standard contains tables with allowable values relating to different editions of IEC Enter IEC IEC Medical Electrical Equipment—recurrent test and test after repair of medical electrical equipment—defines the requirements of ensuring the in-service electrical safety of EM equipment and systems. Sentinel Event Alert 60—Developing a reporting culture: