Instruction manual of EZM Programmable Counter consists of two main . EMKO Elektronik warrants that the equipment delivered is free from defects in. BUT it’s a negative number and I cannot figure out what to do? HELP PLEASE? EMKO ESM Temperature Controller-img_jpg. ESM, ESM, ESM, ESM, ESM ESM 8 mm. Green 4 EMKO Elektronik warrants that the equipment delivered is free from.
|Published (Last):||7 January 2006|
|PDF File Size:||8.70 Mb|
|ePub File Size:||5.61 Mb|
|Price:||Free* [*Free Regsitration Required]|
I am technically challenged but could tell me how to test the thermocouple exactly?
Can you reach the thermocouple inside of the drum? If you can may not be possible you could unplug the thermocouple from the controller and plug into the handheld device.
When you said ” check is to see if the thermocouple type is set properly in the controller” Results 1 to 9 of 9. In my scenario I could route my cabling to this device when I was having issues Hello Fellow Roasters, we just received our Toper 15 kilo roaster from Turkey and could not be more excited to start roasting. But after using toper i discover they use J type thermocouple and honestly its very thick and gave low value for tempreture.
I have the pre-set temp at but the top number should read the current temp of the drum Steve Green Mill City Roasters. One you know esm-443 type you have you need to go into the controllers settings to see if it’s setup for that type of thermocouple.
Your thermocouple is likely disconnected or has a broken wire. Some temp controllers indicate a bad or disconnected thermocouple differently.
I would hope that the manufacturer would tell you this. Can you shoot me a link or something to a hand held device I could purchase to use? It’s odd that its showing negative You may fix it for sure. If you take a look at the attached picture All times are GMT Hi emmko I just read your post about toper drum tempreture.
HTA Instrumentation (P) Ltd.
They have excellent customer service. Originally Posted by fivestarcoffeeroasters. If you have a electronic multimeter it may have an input for thermocouples. Obviously be careful you turn on your drum and its cold Brand new then contact Toper. Its the wiring problem you have to opposite the wire in behind the temperature controller. You can test a thermocouple with a multimeter, it should read a few ohms, I would suggest a disconnected probe or bad connection at the controller?
Charlie If you are afraid of failure or losing money, quit while you are ahead. I use different thin thermocuple connected with Phidget Yes the bottom value is setting the maximum temp it will reach before the gas gets cut off.
So first thing you need to do is check the type of thermocouple you have BUT it’s a negative number and I cannot figure out what to do?
EMKO ESM Temperature Controller
I emkk assume the bottom value that you are setting is the maximum temperature it will reach before cutting off the gas valve fire safety feature. How do I check this?
Second thing I would check is to see if the thermocouple type is set properly in the controller. The time now is The one in my older emk flashes the display at loss of continuity, the phidget shows a negative number.
Thanks for your help! I own toper cafemino.